The SATLOOK NIT COLOR is a swedish made SAT-TV instrument and
spectrum-analyzer.The unit is made for exact alignment and adjustment of
satellite-dishes. It is intended for professional use when high accuracy and
precise information are needed.
It´s easily operated without a lot of unnecessary buttons and knobs. The
basic functions are easy to get a hold on and takes only minutes to learn. The
instrument is provided with a 5″ 16:9 LCD-monitor which either shows “Free to
air” SAT-TV-channel (Digital or Analog), the frequency spectrum 920-2150 MHz (or
parts thereof) alt. Digital data. Menus/help-displays are shown on the LCDscreen
(64×128) beside the monitor.
“Free to air” Digital (DVB-S, QPSK) and Analog SAT-TV channels can be tuned
in and viewed upon (Multistandard Video PAL, NTSC and
Audiofrequencies between 5.5 – 8.5 MHz can be listened to.
The spectrum can be expanded (zoom in) for correct adjustment of the
polarisation (“cross-polarisation”).The instrument has very high resolution and
accuracy. It presents measured data ±2 dB (at around 20° C). SATLOOK NIT COLOR
presents, under one of the sub menus, digital information like BER (bit error
rate), constellationdiagram (QPSK) and S/N (signal/noise ratio).
SATLOOK NIT COLOR in the bitstream (NIT = Network Information Table). The NIT
also contains info about the transponders TV and Radio channels.
SATLOOK NIT COLOR can handle a lot of memory-positions and both
spectrum-pictures and certain frequencies can be saved (both analog and digital
transponders). The polarisation of the LNB is switchable 13V/18V and the Hi-Lo
band with 22 kHz-tone. The DiSEqC-function controls all DiSEqC-accessories (like
switches, LNB´s and actuators). The power of the instrument is supplied by a
built in and rechargeable Li-lon battery. Even though the instrument has a lot
of functions it is still very light and flexible.
Input frequency: 920-2150MHz, easily reduced to250 MHz (max zoom in).
Frequency display: Yes, IF default. All standard LNB L.O. can be
Min level in: About 35 dBμV (noise).
Max level in: About 90
Attenuation: 15 dB attenuator on/off.
(analog) dB on LCD-monitor. Pitch-tone.
Accuracy: ±2 dB (at
Display of signallevel: (Digtal) S/N (signal/noise-ratio) BER (bit
error rate) Constellation (QPSK)
Satellite-identification: Yes, by reading
out the NIT in the bitstream. Info about the TV and Radio channels.
standard: Multi TV/Audio (PAL, NTSC, SECAM).
KU- C-band: Yes,
Audio bandwidth : Adjustable between 5.5 and 8.5 MHz
impedance: 75 Ohm, BNC-con.
Picture-screen: 5” 16:9 TFT-LCD monitor
On LCD 64×128.
Memory: 100 spectrum pictures can be stored with name. Stored
spectrum can be mixed for easy identification of
PC-connection: Yes, RS232-output
22 kHz tone: Yes, on/off.
DiSEqC: Yes, all 1.0 and 1.1. Also
DiSEqC actuator: Built in positioner for DiSEqC 1.2,
SatScan and SatSelect.
Battery: Rechargable 12v, 3.5
Operational: About 1 hour on a fully charged battery.
About 3 kg incl.battery.